Permanent URL to this publication: http://dx.doi.org/10.5167/uzh-29065
|PDF - Registered users only|
Objective: To determine the accuracy of and to establish reference values for a rebound tonometer (Tonovet ) in normal feline eyes, to compare it with an applanation
tonometer (Tonopen Vet ) and to evaluate the effect of topical anesthesia on rebound tonometry.
Procedures: Six enucleated eyes were used to compare both tonometers with direct manometry. Intraocular pressure (IOP) was measured in 100 cats to establish
reference values for rebound tonometry. Of these, 22 cats were used to compare rebound tonometry with and without topical anesthesia and 33 cats to compare the rebound and applanation tonometers. All evaluated eyes were free of ocular disease.
Results: Both tonometers correlated well with direct manometry. The best agreement with the rebound tonometer was achieved between 25–50 mmHg. The applanation
tonometer was accurate at pressures between 0 and 30 mmHg. The mean IOP in clinically normal cats was 20.74 mmHg with the rebound tonometer and 18.4 mmHg with the applanation tonometer. Topical anesthesia did not significantly affect
Conclusions: As the rebound tonometer correlated well with direct manometry in the clinically important pressure range and was well tolerated by cats, it appears suitable
for glaucoma diagnosis. The mean IOP obtained with the rebound tonometer was 2–3 mmHg higher than that measured with the applanation tonometer. This difference is within clinically acceptable limits, but indicates that the same type of tonometer should be used in follow-up examinations in a given cat.
|Item Type:||Journal Article, refereed, original work|
|Communities & Collections:||05 Vetsuisse Faculty > Veterinary Clinic > Department of Farm Animals > Division of Herd Health|
05 Vetsuisse Faculty > Veterinary Clinic > Equine Department > Division of Ophthalmology
|DDC:||570 Life sciences; biology|
|Deposited On:||01 Feb 2010 15:16|
|Last Modified:||23 Nov 2012 14:39|
|Additional Information:||The definitive version is available at www.blackwell-synergy.com|
Users (please log in): suggest update or correction for this item
Repository Staff Only: item control page