Permanent URL to this publication: http://dx.doi.org/10.5167/uzh-31231
Beutel, J; Gruber, S; Hasler, A; Lim, R; Meier, A; Plessl, C; Talzi, I; Thiele, L; Tschudin, C; Woehrle, M; Yuecel, M (2009). PermaDAQ: a scientific instrument for precision sensing and data recovery in environmental extremes. In: Institute of Electrical and Electronics Engineers Corporation, IEEE. International Conference on Information Processing in Sensor Networks, 2009: IPSN 2009, San Francisco, California, USA, 13 - 16 April 2009. Piscataway, NJ, US, 265-276 . ISBN 978-1-4244-5108-1.
PDF - Registered users only
The PermaSense project has set the ambitious goal of gathering real-time environmental data for high-moun- tain permafrost in unattended operation over multiple years. This paper discusses the specialized sensing and data recovery architecture tailored to meet the preci- sion, reliability and durability requirements of scientists utilizing the data for model validation. We present a custom sensor interface board including specialized sen- sors and redundancy features for end-to-end data valida- tion. Aspects of high-quality data acquisition, design for reliability by strict separation of operating phases and analysis of energy efficiency are discussed. The system integration using the Dozer protocol scheme achieves a best-in-class average power consumption of 148μA con- siderably exceeding the lifetime requirement.
|Item Type:||Book Section, refereed, original work|
|Communities & Collections:||07 Faculty of Science > Institute of Geography|
|DDC:||910 Geography & travel|
|Deposited On:||19 Feb 2010 17:15|
|Last Modified:||27 Feb 2014 10:01|
|Additional Information:||International Conference on Information Processing in Sensor Networks, San Francisco, Calif., 2009.04.13-16. - © 2009 IEEE. Personal use of this material is permitted. However, permission to reprint/republish this material for advertising or promotional purposes or for creating new collective works for resale or redistribution to servers or lists, or to reuse any copyrighted component of this work in other works must be obtained from the IEEE|
|Citations:||Web of Science®. Times Cited: 1|
Users (please log in): suggest update or correction for this item
Repository Staff Only: item control page