In this publication we present an improvement to our previously introduced vertical flow microarray, the FoRe array, which capitalizes on the fusion of immunofiltration and densely packed micron test sites. Filtering samples through individual microarray spots allows us to rapidly analyze dilute samples with high-throughput and high signal-to-noise. Unlike other flowthrough microarrays, in the FoRe design samples are injected into micron channels and sequentially exposed to different targets. This arrangement makes it possible to increase the sensitivity of the microarray by simply increasing the sample volume or to rapidly reconcentrate samples after preprocessing steps dilute the analyte. Here we present a new inlet system which allows us to increase the analyzed sample volume without compromising the micron spot size and dense layout. We combined this with a model assay to demonstrate that the device is sensitive to the amount of antigen, and as a result, sample volume directly correlates to sensitivity. We introduced a simple technique for analysis of blood, which previously clogged the nanometer-sized pores, requiring only microliter volumes expected from an infant heel prick. A drop of blood is mixed with buffer to separate the plasma before reconcentrating the sample on the microarray spot. We demonstrated the success of this procedure by spiking TNF-α into blood and achieved a limit of detection of 18 pM. Compared to traditional protein microarrays, the FoRe array is still inexpensive, customizable, and simple to use, and thanks to these improvements has a broad range of applications from small animal studies to environmental monitoring.